Instructions for monitoring and analysis of SIDIS runs during shift
Instructions for SIDIS Experiment
1. Keep an eye on the Hall C Scaler Data GUI and make sure the rates with beam ON are steady
2. In the middle of the run, check online plots for the detectors
a. Login to cdaql1 machine and go to hallc_replay directory if you are not there already:
> ssh cdaql1 > go_analysis
b. Update the file " standard.Kinematics "
path : /net/cdaqfs/home/cdaq/hallc-online/hallc_replay/DBASE/COIN/STD/standard.kinematics
c. Run scripts
for SHMS detector check: ./run_coin_shms.sh <run_number>
for HMS detector check: ./run_coin_hms.sh <run_number>
3. After the full run, check the Good events and Update the run_list wiki
a. Make sure the standard.Kinematics file was updated correctly in step 2 above.
b. For a quick report on the charge collected in the run Analyze the full run with :
./run_charge_sidis.sh <run_number> and record the total charge in the run-list wiki.
c. Analyze the full run with :
./run_full_sidis.sh <run_number> and record the total charge of the run in the run-list wiki.
d. Run the script and record counts to run_list wiki
./UTIL_SIDIS/run_beancounter.sh <run_number> <target> <target> = d or h